The Solinst Model 102 Water Level Indicator is designed to measure groundwater levels, especially in small diameter tubes and piezometers, or where a flexible assembly is needed to get past down hole instrumentation.
The Model 102 Water Level Meter uses a choice of two small diameter probe designs, attached to a narrow coaxial cable, fitted on a sturdy ergonomically designed reel. A standard 9 volt battery, housed in an easy-access battery drawer, powers the Water Level Meter.
When the probe enters water the electrical circuit is completed, sending a signal back to the reel where a light and clearly audible buzzer are activated. The water level is then determined by taking a reading directly from the cable at the top of the well casing or borehole.
A sensitivity control allows the buzzer to be scaled back or turned off while in cascading water, and ensures a clear signal in both high and low conductivity conditions.
Laser Marked Cable
The coaxial cable has a durable polyethylene jacket with permanent markings precisely laser etched on the cable every 1/100 ft. or each millimeter.
The cable reels smoothly, remains flexible, and comes in lengths from 100 ft. – 1000 ft. (30 m – 300 m).
A braided copper outer conductor gives flexibility and a stranded stainless steel central conductor gives strength. It is easy to repair and splice.
Laser Markings Each 1/100 ft. or Each Millimeter
Model 102 Water Level Indicator Probes are designed for flexibility. The probe tips use a recessed design to minimize false signals in cascading water.
P10: 3/8″ dia. x 2.75″ long (10 mm x 70 mm) stainless steel with 10 stainless steel weights, weighing 6.14 oz (174 g). This probe is ideal for the majority of water level measurement applications. The flexible weight assembly allows easy well access and keeps the cable hanging straight in monitoring wells.
P4: 0.157″ dia. x 1.5″ long (4 mm x 38 mm) stainless steel weighing 0.35 oz (10 g), with no weights. It is ideal for shallow applications and for accessing very narrow diameters, especially the channels of the Solinst CMT System, Drive- Point Piezometers or narrow open tubes.